If you have not been trained on how to perform factory setup you can easily render the dispenser inoperative and/or inaccurate!
Get me out of here...
This is done using the CUI AMT Viewpoint software. Save the point beyond the set screw, then insert the set screw to limit travel.
This will set the home offset of both arm axes. Insert the jig and roughly align with X axis.
The park position is the position inside the "hook". The arms must be pushed all the way back into the hook towards the body of the dispenser. Do not teach the position at the front of the hook or it could fail to park since it can clip the opening of the metal hook.
(NB - SerialNumber not stored on device currently..., shown on register only... however is in DB against SerialID)
This will create default calibration tables and set default calibration to be active for both valves.
Ensure any new valves from Gyger are registered into the DB. We track Gyger valve serial numbers so that we can check valve type at a later time.
To search for a valve you must
enter both the serial number and the valve series.
These are written on the valve.
Motion testing will cycle the D2 through a sequence of dispense protocols of various speeds so that we can test the XYZ motion. (Remove the valves for this test or ensure valves current disabled)
Dispense testing will dispense onto a balance and perform a sequence of tare balance/dispense/read balance to ensure that each valve channel is working as it should.